Face-to-face consultations are a key part of patient care. The communication which happens between the health care practitioner and the patient is vital for reassurance and conveying empathy. Even though technology has progressed so far as to provide assistive interpretations,Read More
KUDUwave now has the functionality to conduct audiometric tests at specific frequencies for HSE screening and formulates the results according to OHSAS 18001, categorizing patients quickly and correctly. By following the steps below, you will be able to configure theRead More
Patient response statistics allow the clinician to scrutinize the reliability of the hearing assessment through the responses of the patient. The KUDUwave counts the number of times the response button is clicked during the hearing assessment. It also is ableRead More
Find out the latest updates and features of the new KUDUwave 5 software KUDUwave 5 Change log
X-check (Cross Check) is a digital calibration verification tool built in your KUDUwave. X-check replaces daily biological verification with quick and accurate, digital checks to ensure quality testing. Every audiologist, audiometrist or health care professional who performs hearing tests mayRead More
A laptop with the following minimum specifications is recommended to partner with your KUDUwave.
Smart folders are a new software feature that makes file management and testing a dream for your industry.
To conduct tele-audiology, eMoyo suggests using the popular app TeamViewer.
The assessment of speech in noise provides important diagnostic information which the pure tone audiogram cannot reliably predict. The KUDUwave provides speech in noise testing through its integration with QuickSIN. To access QuickSIN, add a patient or select an alreadyRead More
OneDrive is part of the Microsoft Office suite as an online service. It is a file hosting and synchronization service, well suited as a cloud storage solution for anyone with a Microsoft account.
It is important to note that ordinary audiometers whether they are screening audiometers or diagnostic, do not attenuate sound. The combination of audiometric booths and standard audiometry headsets is what is relied upon to generate the sound blocking required andRead More
Compile a comprehensive report for a referral with a short questionnaire, by summarizing the quality of the data collected and interpretation of the hearing loss detected during screening. This reporting feature allows you to complete a shortlist of questions regardingRead More
Change the view setting in KUDUwave 5 to see calculated HSE results, according to the HSE categorization scheme. How To Get Started Begin by capturing all patient information including age and gender. Conduct a normal audiometery screening test. Ensure the automatedRead More
Using this smart folder type simplifies the method of testing to identify early hearing loss and to calculate measures for compensation. The KUDUwave uses multiple criteria (OSHA, SANS, WHO etc.) to develop testing protocols and assistive interpretations of results. The KUDUwave softwareRead More
This folder specifically caters for the MDR-TB set-up in South Africa, but can be applied to other projects across the world. The KUDUwave uses multiple criteria (OSHA, SANS, WHO etc.) to develop testing protocols and assistive interpretations of results. The KUDUwave softwareRead More
How to use the Humanitarian Screening Smart Folder This Smart Folder is designed for efficient and effective mass screening. The quick automatic test sweeps through 5 frequencies between 25dB HL and 90dB HL. The KUDUwave uses multiple criteria (OSHA, SANS,Read More
Upgrading your KUDUwave software and transferring patient data is made simple with a few straight forward steps. Upgrading from eMoyo Live Step 1: BackupTo backup your data, go to the settings page and click on the first option: “Backup…” . NextRead More
A complete audiological test is automatically given the significance ‘none’. To change this to be more specific about which test was performed (i.e. baseline/monitoring/exit), you must change the significance. Where is the test significance? In the Patient Profile pane, youRead More
Here are three easy steps to follow to download and install the latest version of KUDUwave™ Software. 1. Download You can download the latest software directly from this link. Just fill in your details and click download at the end.
Look professional with a personalized letterhead on all your patient records generated from your KUDUwave. To include your personalized letterhead on patient audiograms, you or your graphic designer will need to create a graphic that can be uploaded into yourRead More
Discover where to change your default printer settings. Here are step-by-step instructions to troubleshoot printing audiograms from your KUDUwave software. If you have recently updated your laptop or windows operating system, you may have encountered an error message with yourRead More
Here’s how to do it: Step 1: Launch your KUDUwave software and click on the X-check icon in the bottom left hand corner of your home screen. The X-check start page will appear:
X-check (Cross Check) verifies the calibration of your KUDUwave digitally. Like biological checks, X-check does not replace the need for regular calibrations as per the need for regular calibrations. X-check is a new feature that comes standard with all KUDUwave™ Prime, KUDUwave™ Plus and KUDUwave™ Pro devicesRead More
“Access other computers or allow another user to access your computer securely over the Internet.” You will need to have installed Google Chrome on each PC’s to use this application.